Home / Rae Systems by Honeywell / Hydrogen Sulfide Replacement Sensor for AreaRae / MultiRae / ToxiRae Pro Hydrogen Sulfide Replacement Sensor for AreaRae / MultiRae / ToxiRae Pro$271.00 Ships directly from the manufacturer, shipping times may vary. Call for availability Hydrogen Sulfide Replacement Sensor for AreaRae / MultiRae / ToxiRae Pro quantity Add to cart SKU: C03-0907-001 Categories: Rae Systems by Honeywell, Store Tags: MulitRae Sensors, ToxiRae Sensors Description Additional information Product Enquiry Description Genuine Hydrogen Sulfide Replacement Sensor for AreaRae / MultiRae / ToxiRae Pro gas detectors from Rae Systems by Honeywell. All specifications have been verified under the following environmental conditions: Temperature: 68°F Relative Humidity (non-condensing): 50% Ambient Pressure: 1 atm (1,013 mbar) Refer to the Technical Note TN-114 for a glossary of terms used on this page Hydrogen Sulfide Replacement Sensor Specifications Sensor Type Electrochemical Range 0-100 ppm Resolution 0.1 ppm Drift <2% signal/month Bias/Equilibration No bias/10 minutes after installation Storage Life 6 months in sealed container Operating Life 2 years in air Calibration Gas 10 ppm Hydrogen Sulfide in Nitrogen Temperature Range -4°F to 122°F Storage Temperature Range 32°F to 68°F Pressure Range Atmospheric ±10% Operating Humidity 15 to 90% rH non-condensing Warranty 2 years from date of shipment Cross Sensitivity Data Gas Concentration Response Carbon Monoxide 300 ppm ≤1.5 ppm Carbon Disulfide 100 ppm 0 ppm Ethyl Sulfide 100 ppm 10 ppm2 Ethylene 100 ppm ≤0.2 ppm Hydrogen 3,000 ppm 0 ppm Hydrogen Chloride 10 ppm 0 ppm Hydrogen Cyanide 10 ppm 0 ppm Isobutylene 100 ppm 0 ppm Methyl Mercaptan 5 ppm about 2 ppm Methyl Sulfide 100 ppm 9 ppm Ammonia 50 ppm 0 ppm Nitric Oxide 35 ppm <0.7 ppm Nitrogen Dioxide 5 ppm about -1 ppm1 Phosphine 5 ppm about 4 ppm Sulfur Dioxide 5 ppm about 1 ppm Toluene 10,000 ppm 0 ppm2 Turpentine 3,000 ppm about 70 ppm2 NOTE: High levels of polar organic compounds including alcohols, ketones and amines give a negative response. 1 – CAUTION! Negative cross sensitivities may cause the sensor to produce lower readings than the true concentration of gas in ambient air 2 – Estimated based on date from similar sensors Downloads Technical Note: TN-114 – Sensor Specifications and Cross Sensitivities for Hydrogen Sulfide Replacement Sensor Disclaimer: Technical Note TN-114 presents specifications, cross sensitivities and calibration information on select Rae Systems by Honeywell sensors. All specifications presented in this Technical Note reflect the performance of standalone sensors. For instrument specifications, please refer to product datasheets and manuals. Actual sensor characteristics may differ when the sensor is installed in different instruments. As sensor performance may change over time, specifications provided are for brand new sensors. Additional information Weight .5 lbs Product Enquiry Name Email address Enquiry Related products RaeLink3 Mesh Wireless Modem with Integrated GPS Ships from Manufacturer, shipping times can vary MultiRAE Benzene with Benzene Specific Measurement Ships directly from the manufacturer, shipping times may vary. QRae 3 Wireless 4 Gas Detector from Rae Systems Ships from Manufacturer, shipping times can vary Rae Systems MultiRae Pro Wireless Radiation & Chemical Detector Ships directly from the manufacturer, shipping times may vary.